Writer： admin Time：2021-06-05 14:42 Browse：℃
High temperature sterilization pot add appropriate amount of water in the outer pot, put the articles to be sterilized into the inner pot, cover the pot cover and tighten the screw symmetrically. When the pressure gauge pointer reaches 33.78kpa, open the vent valve to discharge the cold air. At this time, the pressure gauge pointer drops. When the pointer drops to zero, close the vent valve to continue heating, the steam in the pot increases, and the pressure gauge pointer rises again. When the pressure in the pot increases to the required pressure, the firepower decreases, Maintain the steam pressure for a certain period of time, then cut off the power or fire of the sterilizer, let it cool naturally, and then slowly open the exhaust valve to remove the residual gas, and then open the cover to take things.
The control mode adopts microcomputer intelligent automatic control to control sterilization pressure, temperature and time; Over temperature automatic protection device: over the set temperature, automatically cut off the heating power supply; Door safety interlock device: there is pressure in the inner cavity, the door cover can not be opened, so it is a patented device; Low water level alarm: it can automatically cut off the power supply in case of water shortage, sound and light alarm, imported water cut-off detection device; Leakage protection: equipped with leakage protection device; Temperature dynamic digital display, send out the end signal after sterilization; Heating, sterilization, exhaust, drying process automatic control, without manual supervision.
When the time required for sterilization is up, cut off the power or turn off the gas to let the temperature in the sterilization pot drop naturally. When the pressure of the pressure gauge drops to 0, open the exhaust valve, loosen the bolt, open the cover and take out the sterilized articles. If the pressure does not drop to 0, open the vent valve, the pressure in the pot will drop suddenly, and the medium in the container will rush out of the flask or tube mouth due to the imbalance of internal and external pressure, resulting in the contamination of the medium by the cotton plug.
Add: 10047 Xinxing Road, Xinxing Town, Zhucheng City, Shandong Province, China